Historical Standard: ASTM A912/A912M-04 Standard Test Method for Alternating-Current Magnetic Properties of Amorphous Materials at Power Frequencies Using Wattmeter-Ammeter-Voltmeter Method with Toroidal Specimens
SUPERSEDED (see Active link, below)
ASTM A912 / A912M
1.1 This test method covers tests for various magnetic properties of amorphous materials at power frequencies [25 to 400 Hz] using a toroidal test transformer. The term toroidal test transformer is used to describe the test device reserving the term specimen to refer to the material used in the test. The test specimen consists of toroidally wound flat strip.
1.2 This test method covers the determination of core loss, exciting power, rms and peak exciting current, several types of ac permeability, and related properties under ac magnetization at moderate and high inductions at power frequencies [25 to 70 Hz].
1.3 With proper instrumentation and specimen preparation, this test method is acceptable for measurements at frequencies from 5 Hz to 100 kHz. Proper instrumentation implies that all test instruments have the required frequency bandwidth. Also see Annex A2.
1.4 This test method also provides procedures for calculating impedance permeability from measured values of rms exciting current and for calculating ac peak permeability from measured peak values of total exciting current at magnetizing forces up to about 10 Oe [796 A/m].
1.5 Explanation of symbols and brief definitions appear in the text of this test method. The official symbols and definitions are listed in Terminology A 340.
1.6 This test method shall be used in conjunction with Practice A 34/A 34M.
1.7 The values and equations stated in customary (cgs-emu and inch-pound) units or SI units are to be regarded separately as standard. Within this standard, SI units are shown in brackets. The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other. Combining values from the two systems may result in nonconformance with this standard.
1.8 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
A34/A34M Practice for Sampling and Procurement Testing of Magnetic Materials
A340 Terminology of Symbols and Definitions Relating to Magnetic Testing
A343/A343M Test Method for Alternating-Current Magnetic Properties of Materials at Power Frequencies Using Wattmeter-Ammeter-Voltmeter Method and 25-cm Epstein Test Frame
A901 Specification for Amorphous Magnetic Core Alloys, Semi-Processed Types
C693 Test Method for Density of Glass by Buoyancy
amorphous; core loss; exciting force; exciting power; impedance permeability; peak permeability; specific core loss; specific exciting power; tape wound core; toroidal test specimen; toroidal test transformer; waveform distortion: A-C (alternating-current) magnetic properties; Alternating-current permeability; Amorphous magnetic material; Core loss--alternating-current; Exciting force/power; Impedance permeability; Magnetic properties; Peak permeability; Specific core loss; Tape wound core; Toroidal transformer; Waveform distortion;
ICS Number Code 77.040.99 (Other methods of testing metals)